I found from a dismantled lab sell a Fluke PM I never heard anything about Fluke analog scopes. plus the name looks like a Philips. Find great deals for Fluke PM Analog Oscilloscope. FLUKE PM , PM , PM, PM ANALOG OSCILLOSCOPE SERVICE MANUAL. Find great deals for Fluke PM Oscilloscope 4 Channels mhz No Cable. Shop with confidence on eBay!.
|Published (Last):||25 September 2005|
|PDF File Size:||12.65 Mb|
|ePub File Size:||10.22 Mb|
|Price:||Free* [*Free Regsitration Required]|
This is displayed on the CRT with a message: Put CH1 in 0. Page Set CH1 to 0. Circuit diagram of Y-out unit Rough handling or scratching can cause the CRT to implode.
Fluke PM Oscilloscope 4 Channels mhz No Cable | eBay
However, many of the standard electronic components can be obtained from other local suppliers. When instrument is switched off or during mains failure. Adjust the input signal to an amplitude of exactly 6 divisions. Printed circuit board lay-out of Y-out unit After exchange it must be filled with operating software and calibration data.
Page Verify that the jitter of the DTB is not more than 0. On the power supply unit there are many parts that carry dangerous high voltages.
Fluke PM3082 Service Manual
Page of Go. The x1 is switched by control signal PA1X1. Use the internal termination when available. Don’t have an account?
DTB jitter in starts 1 part of 2. For installation, reverse the sequence. These names make it easier to trace a signal going from one circuit diagram to flu,e.
CHIP For the MHz version the corresponding circuit diagram should be used. TV pattern generator with video output PM test set-up: Configuration of mmicroprocessor control part Printed circuit board lay-out of p,3082 trigger vluke To print the manual completely, please, download it.
Error Calibration step Input Control signal Output When ready, the oscilloscope is fine tuned to optimal accuracy. Otherwise the component is located on the side with small components.
Page – Maximum phase shift between horizontal a Remove all superfluous soldering material. The oscillation principle is now explained with the simplified diagram in the figure. Press the CAL key for a few seconds to start the autocal procedure.
Use desolder braided wire; ordering code: The output circuits are protected against overload by the under voltage protection. There are flue vertical channels that are mainly identical.
The output signal from the delay line is applied to the input pins 6 and 9 of amplifier IC D Page – Z-MOD sensitivity Page DC coupled mode. The input circuit is formed by IC D Verify that a horizontal line of 4. Manual calibration data are protected by a keyword and a seal and pm33082 be changed by qualified personnel only.
Carefully unsolder the soldering tags of a semi-conductor one after the other. The external trigger requires a small printed vluke board, a rear panel BNC socket and a coaxial interconnection cable.
Synchronization is achieved via SCL. To reduce the rise of electric shock donot perform any servicing other than that contained in the operating In a sequence. Page bcc U p BCC: The principle of the time base is that a capacitor is charged with a constant current. Don’t show me this message again.
Each data block sent by the microprocessor is preceeded by an address on which the device can respond. The output signals are measured on the X-deflection plates of the CRT.